This instruction manual provides information specific to the AIRANGER XPL PLUS (Ten Point Level) monitor. When references are made to other Milltronics. AiRanger XPL Plus. * Program range is defined as the empty distance to the face of the transducer (P) plus any range extension (P). **EMC performance. AiRanger XPL Plus. Ten-Point Ultrasonic Level Measurement System. Part 1. General. Scope. A. This section describes the requirements for an ultrasonic.
|Published (Last):||3 April 2005|
|PDF File Size:||12.65 Mb|
|ePub File Size:||8.24 Mb|
|Price:||Free* [*Free Regsitration Required]|
Remove the 4 board B mounting screws outer corners and remove the circuit board assembly. Scanning From this graph, determine which Universal Volume calculation will provide the best results, and select the level vs volume breakpoints to be entered 32 max.
Before using this feature, verify the: Use these features to record and save a total of up to 10 Echo profiles, initiated manually Por automatically P et al. Select “short” to have only short shots fired if the Echo Confidence P produced by a short shot exceeds the short Confidence Threshold P Page 90 P Tank shape Enter “1” for vertical cylinder with flat bottom.
Milltronics airanger XPL plus Instruction Manual
Data messages are continuously transmitted at regular intervals no poll is required at baud. Resume making your offer if the page doesn’t update immediately.
An item that has been previously used. Learn More – opens in a new window or tab Any international postage and import charges are paid in part to Pitney Bowes Inc. Access the Pluus P parameter. Measurement Parameters P to Plux Keypad Parameter Value in memory complete a parameter reset. Operation Point scanned independent from the Point Number display.
MILLTRONICS AIRANGER XPL PLUS INSTRUCTION MANUAL Pdf Download.
Page 53 Alternatively, if the amount of Offset Correction required is known and performing an Offset Calibration is not desiredenter the amount to be added to the Reading before display. This feature is preset to 0. Enter the amount in ms of long shot Echo Profile smoothing required airnager eliminate jagged peaks in the Echo Profile. Introduction After reading this introduction and completing the x;l installation, first time system installers may wish to use the AiRanger XPL Plus Quick Start Guide for step-by-step start up instructions.
No entry required, preset for metres.
Milltronics AiRanger XPL Plus Instruction Manual
From this graph, determine which Universal Volume calculation will provide the best results, and select the level vs volume breakpoints to be entered 32 max. If “FAIL” is displayed, repeat the test. By pressing ‘print’ button you will print only current iaranger. If the material presents a flat surface perpendicular to the transducer beam, select liquid.
To select an independent Relay Failsafe option value: A negative rate indicates the vessel is emptying.
Following a Global Master Reset, complete reprogramming is required. P 4 mA TRIM Use this feature in combination with 20 mA Trim if the device connected to the mA output displayed is out of calibration, yet device recalibration is impractical.
Space or Remaining Vessel Capacity Measurement for a single vessel. If the peak value jumps wildly, verify the transducer cables are run in grounded metal conduit; Not all components or their maximum quantity may be required. Buy it now – Add to Watch list Added to your Watch list. Page 68 Extend Near Blanking to overcome p,us difficulties which cannot be corrected by transducer location, mounting, or aiming.
To print the manual completely, please, download it.
All associated data is supplied to the Peripheral Communications terminals 42 Note: Add to Watch list. Distance Measurement less than 60 m.
Enhancement Parameters Scope Displays, P Alternatively, if the amount of Offset Correction required is known and performing an Offset Calibration is not desiredenter the amount to be added to the Reading before display. Add to basket. This value is automatically adjusted when an Offset Calibration is performed.
Siemens SITRANS LU02 Dual Point Ultrasonic Level Transmitter (Milltronics AiRanger DPL Plus)
Enter text from picture: Seller assumes all responsibility for this listing. Operation is similar to Space Measurement operation except that measurements are referenced to the transducer face.
By pressing ‘print’ button you will print only current page. Page Extend Near Blanking P to a distance from the transducer face to just beyond the distance represented by the Reading. Enter the number of long shots to be fired and results averaged per transmit pulse. Page 70 Profile Pointer. Follow qiranger grounding procedures.